Rev. 5 - 11/08
Copyright© 1997-2008, Alpha Communications®, All Rights Reserved
42 Central Drive • Farmingdale NY 11735-1202 • Phone (631) 777-5500 • Fax (631) 777-5599
TOLL-FREE 1-800-666-4800 • Web: www.AlphaCommunications.com • Email: email@example.com
The RY351B SideCom® adaptor is designed for use
with the IR311C, IR312C, IR319B and/or IR320B
patient stations, and a HILL-ROM® SideCom unit.
When used with the above patient stations and the
HILL-ROM® SideCom® unit the RY351B adaptor
will provide all the necessary circuitry for a HILL-
ROM® SideCom® unit to control all functions nor-
mally controlled by a pillow speaker. This includes
but is not limited to nurse call and patient communi-
cations. Note: There is one RY351B SideCom®
adaptor required per SideCom® unit.
• Provides for the following functions from a Hill-Rom®
Hands-free two-way communication
• Fully supervised
• Ramp lock plug-in connectors
Height: 1.25" (31.8mm)
Width: 2.00" (50.8mm)
Length: 4.25" (108mm)
Circuit board mounted on a
rugged aluminum mounting
Reliable ramp lock plug-ins.
NOTE: The RY351B comes with one (1) dummy
plug (for the IR311C or IR312C only) and re-
quires a model PL301 blank plate and IH358
electrical box with IH357 plaster ring. It is used
with the Hill-Rom® #P376 Wall Plate adaptor
unit (provied by Hill-Rom). When used with the
IR319B or IR320B patient station, you require
one or two separate JK401 dummy plgs, respec-
Due to continuous product improvement, all colors, sizes, materials,
finishes and specifications are subject to change without notice.
ARCHITECTS' AND ENGINEERS'
The interface for the Hill-Rom® SideCom®
bed controls to Alpha Communications®/
Tektone Nurse Call Systems shall be Alpha
Communications®/Tektone RY351B or ap-