Rev. 2 - 08/2010
Copyright© 2009-2010, Alpha Communications®, All Rights Reserved
42 Central Drive • Farmingdale NY 11735-1202 • Phone (631) 777-5500 • Fax (631) 777-5599
TOLL-FREE 1-800-666-4800 • Web: www.AlphaCommunications.com • Email: email@example.com
Due to continuous product improvement, all colors, sizes, materials,
finishes and specifications are subject to change without notice.
ARCHITECTS' AND ENGINEERS'
The NC505ES shall provide remote annunciation of call activity
within the Tek-CARE®500's Wireless Emergency Call System.
It shall function independently of the Tek-CARE®500 master
station computer. It shall provide dry contact outputs which are
Form-C and capable of driving a typical 24-volt dome lamp as used
in the call systems. The NC505ES shall also allow the use of an
external, model TAK-4202 or TAK-4208 dialer, that accepts dry
contact input(s). The NC505ES shall also enable SF501ES, SF501/
2ES and SF502ES transmitters to trigger the external dialer.
All programming shall be non-volatile. The unit shall provide for
easy connection of field wiring. Access to the interior of the case
shall be secured by a built-in tamper switch - for easy connection
to a third-party alarm system. The NC505ES shall be designed for
indoor use only, and shall feature hidden cable connections and
access. The current status of all outputs shall be visually indicated
on the face of the unit.
TekTone® has designed the Tek-CARE®500 to work as a system. Its
NC501ES Master Station Computer has been carefully designed and
matched to work with Tek-CARE®500 wireless and wired peripherals.
TekTone® will not warrant or support components of Tek-CARE®500
that are purchased separately unless the components are add-ons to
previously sold systems. TekTone® will also not warrant or support a
system that is a mix of TekTone® and Non-TekTone® parts, unless
prior written permission has been obtained. In order to track our systems
and to inform facili