Rev. 5 - 02/09
Copyright© 1997-2009, Alpha Communications®, All Rights Reserved
42 Central Drive • Farmingdale NY 11735-1202 • Phone (631) 777-5500 • Fax (631) 777-5599
TOLL-FREE 1-800-666-4800 • Web: www.AlphaCommunications.com • Email: firstname.lastname@example.org
ARCHITECTS' AND ENGINEERS'
Emergency pull/push station(s) shall be Alpha Communi-
cations®/Tektone SF337C or approved equal. Operation
of emergency signals shall be accomplished by a short
pull on the nylon cord or by pulling/pushing down on the
RED lever marked "EMERGENCY".
The red "CALL-PLACED" L.E.D. indicator shall flash to
assure the patient that the call has been placed.
The Emergency station shall be WATERPROOF if mounted
according to manufacturer's instructions. If mounted
correctly, the faceplate shall be sealed from the electronic
Switching mechanism shall consist of a reliable REED
switch with a sealed MAGNET actuator.
The SF337C shall be UL 1069 listed to the latest 4th Edition
standards. Contractor shall observe all local and national
electrical and building codes.
Pull/Push Shower Station
The SF337C may be used in any location where activation of an
emergency signal is required. The design is such that it may be
located anywhere in a shower, bath, or tub room regardless of
The combination of rubber grommetted mounting screws, rubber
mounting gasket, and completely sealed front faceplate make it
WATERPROOF (if properly installed). All electronic compo-
nents are mounted on a circuit board, including an industry-
proven reliable reed switch for circuit initiation. The activating
magnet is mounted on the large pull-down lever which is
Due to the flexibility of its design , the SF337C may be used as
a PULL CORD EMERGENCY STATION, a NON-PULL
CORD EMERGENCY STATION, or a STAFF ASSIST EMER-
GENCY CALL STATION. The pull cord supplied may be used
for emergency call activation or, if so desire