Copyright© 2001, Alpha Communications®, All Rights Reserved
42 Central Drive • Farmingdale NY 11735-1202 • Phone (631) 777-5500 • Fax (631) 777-5599
TOLL-FREE 1-800-666-4800 • Web: www.AlphaCommunications.com • Email: email@example.com
Rev. 1 - 08/01
Master station(s) shall be Alpha Communications®/Tektone
NC304, NC304LCD or approved equal. The master station
wiring shall be supervised. The master station(s) shall include
the necessary signaling circuitry to receive and answer calls
from all patient, staff, and duty stations; and receive calls from
code and emergency stations on the NC300II System. The
master station(s) shall provide for handset or loudspeaker
communication with patients and staff. Function buttons with
(and without) LED's shall be provided to initiate reset opera-
tion, staff emergency request, nurse/aide request, paging opera-
tions, and nurse-follower operations. Additional buttons for
menu-driven programmable functions include: talk, monitor,
capture, page up/down and room priority programming.
For the NC304LCD, incoming calls shall be displayed by room
numbers in plain English on an integral LCD display with a
minimum of 2 lines capable of displaying 40 characters each.
The calls will display in order of time of origination, and in
accordance with established priority. The master station(s)
shall also provide a color VGA output to connect to a standard
VGA monitor (our model# NC313) for color VGA call display.
A parallel port and serial port shall be available for connection
to UL1069® Listed devices.
The user shall be able to access system programming via the
keys on the master station(s) or via the keyboard port, which
will accept any standard PC keyboard (similar to our model#
The master station(s) shall be equipped with an integral handset,
volume control, color-keyed function switches (with and with-
out LED's), audible signal control switches, call LED, busy
LED, 2.50" speaker, electrostatic microphone,